**System performance: V830 - 335 SPA10K**

Usable frequency range | 0 - 3000 Hz |

Rated travel (S=52) | 50.8 mm p-p |

Rated velocity | 2000 mm/s peak |

Rated acceleration (sine operation) | 75 gn peak |

Rated acceleration (random operation) | 60 gn rms |

Rated acceleration (bump operation) | Dependant on pay load |

Rated force (sine operation) | 1000 kgf peak |

Rated force (random operation) | 1000 kgf rms |

Rated force (bump operation) | Dependant on pay load |

Armature fundamental resonance fn | 2300 Hz ±5% |

Armature mass (Hex inserts) | 12.85 kg |

Body mass | 616 kg |

Cross axial stiffness at inserts | 535.7 kgf/mm |

Maximum turning moment | 2868 kgf mm |

Null point to insert | 55 mm |

Internal load support mass limit | 160 kg |

Vibrator isolation system fn | 5 Hz or 10 Hz |

The above list defines the maximum system performance. Some of these values are derived from a combination of the vibrators operating limits and the amplifiers maximum output. For example, if a 5kVA amplifier was used instead of a 10kVA amplifier, the maximum force and acceleration would be reduced, as less current would be available to drive the vibrator.